IniTool.exe is a utility designed to create a configuration file for use with the
M201. Below is a link to a screenshot of how the application looks when it's first loaded|
IniTool.exe automatically starts with default configuration parameters filling all the boxes
that appear on it's main panel. If you already have an M201_SP.INI file that you've created
and have been using, but would like to change a few of the configuration parameters, there is a button which
can be used to read your current IniTool.exe which you can then change as needed.
The various configuration parameters for the M201 can be changed using IniTool.exe and
then use the save and exit button to write the changes to either an existing
M201_SP.INI file if one exists, or create a new one if one does not exist. You can also choose
to just exit without saving any of the changes you may have made. You can also use IniTool.exe
as an easy way to view M201_SP.INI in a format that may be simpler to understand than
would be possible if you were to open M201_SP.INI into a text editor and look at it's
Below, is a few things that may be of interest when using IniTool.exe.
Whenever you change the channels and sub-channels settings, be sure to press the
save the above values to memory now button before proceeding.
A multiplexer must be connected to make use of sub-channels, so if a multiplexer will not be connected,
you should set both the low and high channel to zero for any of the main channels that you will want to
scan during a scan. Also note that whenever a main channel has it's low channel set to 1 and it's high
channel set to 0, that channel will not be scanned.
The minimum scan interval will vary depending on other configuration parameters. After setting all the
other parameters, you can either then select from a range of possible scan intervals, or upon exit,
IniTool.exe will automatically set the interval to the minimum interval based on the other